If you’re still getting errors, some of them are discussed in that same manual. If so, then please attach an NI-Spy capture from the read failure If you have the NI Message 7 of I have attached a picture of the current scan setup to the loop.
|Date Added:||8 November 2004|
|File Size:||10.30 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Thank you in advance. What was the driver that you used in conjunction with the rest of the code? Never mind I found it.
Reture Status should be x0 and Buffer should display instrument info. I am not familiar with the express vi “Write to Measurement File”, but you could separate yours data like this: Message Edited by blawson on Apart from the Keuthley 0 i.
Message 4 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Find more articles tagged with: By creating those folders, kekthley proplem seems fixed.
The data is output directly to the bus instead of storing the data in the instruments internal data buffer. Message 28 of About how long does it take for the values to switch over?
Replace “Read Data Buffer Subset. Most Active Software Boards: Message 2 of 2.
I don’t care about when the measurements were taken, I just a record of the temperature of both channels until I click “Stop. Any ideas would be helpful! Message 3 of 5. Message 29 of Thanks for your support Jean-Marc! The Keithley has a configurable scan rate of Fast, Medium, and Slow. I need to scan between these two channels, correctly determine which channel I’m currently getting a resistance value from, route the resistance to the appropriate conversion equation, then save that output to a file.
Keithley Multimeter/Data Acquisition Sy – Instrument Driver – National Instruments
Ideally the output file would be formatted something like: Only power line cycles, not seconds, works for standard DC measurements. Hi DaleYes ,its very odd for me too. Message 6 of How often does it occur and what steps do you take to reproduce this? If I am reading everything correctly, it looks like the output is just keothley interlaced 1D array. I don’t understand what did i do wrong. In order to resolve this issue, a “Decimate 1D Array” function is used in order to separate values keithly individual channels into separate rows or columns in an array.
Keithley 27XX Switch,Meter,Data Acquisition System
I am attaching the vi and excel file screenshot. Return Status should be x0. I was using the load file.