In this case, it will be much easier to use GetDataEx to place the data directly to a pre-defined integer array. GapInDataStream indicates a gap is detected in the data stream. PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured PreTriggerLength integer specifies the number of scans before the trigger point, 0 to scans can be specified default: Sorry, but the pattern of the square wave cannot be controlled. Fired when the number of new data points are equal to or higher than what is specified in the property of EventLevel. The specified integer must be between 0 and 32,
|Date Added:||21 September 2006|
|File Size:||45.35 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
To enable multiple channels on DI, you must provide the Key before you Start the device. To catch the Minimum reading of a specified channel in the duration of the test from the moment Start method is invoked to the moment of this method is invokedadd to the valid channel count mentioned above. This method must be invoked after Start Applicable Devices: GapInDataStream indicates a gap is detected in the data stream.
Instruments Techno Test inc. | Our products –
The included samples are mostly from VB6. The data will be aligned with the first channel. When Ultimaserial is first loaded, the channel allocation table is 1, 2, 3, Default value is Original channel count is do-194rs To convert the AD Readings, check here. For examples in other programming languages, please visit our classroom. If you need to enable channels with different combination, you can use UltimaSerial.
Count is an integer and defines the number of scans you are requesting. GetDataEx returns waveform data to an integer array as binary data. Invalid COMM port COM port is not available.
Sample out of possible range PGain index changes P rogrammable Gain table. For example, since a DI has 4 possible channels, and you request 10 channels, only one channel will be enabled.
DATAQ Instruments Hardware and Software Manuals
A trigger event will be fired when the trigger condition is met. PreTriggerLength should be zero Applicable Devices: To access the data returned by GetData, you can take a look at the following example If you are using VB.
ChannelCount Reads the number of currently enabled channels Variable: PreTriggerLength should be zero PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured AvailableData: Serial port 1 is the default. When you specify a sample rate, an attempt is made to report data at that sample rate.
DI and DIRS WinDaq Data Acquisition Software Upgrade
Lets you write the number of channels you want to be enabled or read the number of channels that are currently enabled. The order of data points is: To enable a digital channel, you need to use codes like this:.
PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured PreTriggerLength integer specifies the number of scans before the trigger point, 0 to scans can be specified default: You must use the ChannelCount property to enable the appropriate channel before you can invoke this method to get the readings from a specified channel.
Specify the Dataq serial data acquisition device model number you are going to use or find out which one you are currently using. Count, an integer, is the number of data points, not scans, that you are vi-194rs 30, maximum.
Dataq Instruments Di-194rs 4 Channel Data Acquisition Module & Cable
NET, please also check this one out:. NET Class if you are starting a new project on products supported by the class. Since the default gain of 0 is not a valid number, user must specify the range first before using the rate dii-194rs. The default value for gain index is 0.