Note that when configuring for differential inputs, the positive and negative pairs of the signals must be connected as indicated above. Multiple options are entered on the same line separated by semicolons. This continues for the first eight differential inputs. We know there are plenty of differences when it comes to used equipment and quite often, choosing between different pieces is difficult, especially when the equipment is not sitting right in front of you. The following are the available options and valid values for this board. We appreciate your interest in using our InstraView website! The options and their values are not case sensitive for the National Instruments boards.
|Date Added:||1 July 2009|
|File Size:||28.94 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
For example, enter 4: Differential for the first four channels, referenced single-ended for the next four channels. In order to read the differential input values, the input channels must be set to [, ]. When in duty cycle mode, the acceptable range of duty cycles are:. Set the PWM output mode field to the desired output faq.
When adding a timebase block, you must change the Samples in buffer field or you will receive the error “The specified buffer size is not valid. InstraView Request Form To get started: For instance, channels, and are differential. Set the 66062e output channels field to all the PWM output channels that will be used on the board for the current diagram. Click a thumbnail to view larger image: Not just a picture from the manufacturer’s website, but the actual piece of equipment you would receive.
In frequency mode, the acceptable range of frequencies are related to the duty cycle of the signal and are as follows:. See the description of the HIL Initialize block for xaq on these modes. Specify [0, 4, 5] to indicate that lines 0, 4 and 5 are to be configured as dqq inputs.
Ni DAQCard 6062E National Instruments 12-bit PCMCIA DAQ Card
Multiple options are entered on the same line separated by semicolons. Channels, and are referenced-single ended. Stock Quick Quote.
However, attempting to command a value which is out of range would not produce an error. The tenth differential input’s positive line must be connected to channel 17, and negative line to channel In das, an immediate block for the same type of channel can not be used even if they use different resources. The driver would simply saturate the value at the limits.
Hence, a scalar value will apply to all channels specified in the PWM output channels field. Product price, specification, warranty and availability subject to change without notice. Supported with no communication to the hardware.
Submit feedback to Quanser about this page. Hence, a scalar value will apply to all channels specified in the Digital output channels field. We’ll send you an email letting you know exactly when your piece of equipment will be available for viewing Name: Please feel free to contact us at sales artisantg. Specify [0, 4, 5] to indicate channels 0, 4, and 5. The same holds true of the task creation functions.
DAQCard – National Instruments
Please contact 60622e of our Customer Support Specialists if you have any questions about what is included with this equipment or if you require any additional information. The second differential input’s positive line must be connected to channel 1, and negative line to channel 9. Also, a frequency or a period with value of 0 is also unacceptable. Specify [0, 4, daa to indicate that lines 0, 4 and 5 are to be configured as digital outputs.
Manuals, Datasheets, Drivers, Links.